dixell. DEFAULT SETTING VALUES. تا تا تست ه تا تن تا نا م ع. XR60C signals and records temperature alarms, together with their duration and max. View and Download Emerson Dixell XR60C installing and operating instructions online. Digital controller with defrost and fans management. Dixell XR60C. Dixell Srl reserves the right to change the composition of its products, even without or faulty operation send the instrument back to the distributor or to Dixell Srl.
|Published (Last):||25 October 2018|
|PDF File Size:||15.63 Mb|
|ePub File Size:||10.6 Mb|
|Price:||Free* [*Free Regsitration Required]|
Consider the maximum current which can be applied to each relay see Technical Data.
electronic controller DIXELL XR60C 379983
The instrument must not be opened. Don’t have an account?
Add to my manuals Add. To see the last temperature alarm happened; in programming mode it browses the. Each LED function is described in the following table. An additional parameter “FSt” provides the setting of temperature, detected dixeol the evaporator.
This function managed by the Fct parameter is designed to avoid short cycles of fans, that. Check the supply voltage is correct before connecting the instrument.
If the alarm LED is on, an alarm has taken place. Universal variable speed ac drive for induction and permanent magnetmotors pages. The instrument is fully configurable. At the end rx60c defrost dripping time is started, its length is set in the FSt parameter.
Ensure that the wires for probes, loads and the power supply are separated and far enough.
Comments to this Manuals Your Name. Two defrost modes are available through the “tdF” parameter: To dxiell the manual completely, please, download it. Other parameters are used to control the.
This is used to make sure circulation of air only. Digital controller with defrost and fans management. Then the instrument displays the temperature once again.
Emerson Dixell XR60C Installing And Operating Instructions
The instrument shall not be used for purposes different from those described diexll. By pressing ‘print’ button you will print only current page. This manual is part of the product and should be kept near the instrument for zr60c and quick. To display target set dixeell in programming mode it selects a parameter or confirm an.
Dixell Srl reserves the right to change the composition of its products, even without notice. Do not expose to water or moisture: Fit the probe where it is not accessible by the End User. Page of 4 Go. Got it, continue to print. The fan control mode is selected by means of the “FnC” parameter:. Unidrive hs series; unidrive m series. This manual also for: To confirm the operation, the “rSt” message starts blinking and the normal temperature.
On the display the following message is shown: Didell machine integration through communications 50 pages. In case of fault in the thermostat probe the start and stop of the compressor are timed through.
Push and immediately release the SET key or wait for 5 seconds to. Hold the SET key pressed for more than 3s, while the recorded alarm is displayed.
XR60C – AS Catering Supplies
Push and immediately release the SET key: To see the kind of alarm, the max min. The instrument is provided with screw terminal block to connect cables with a cross section up nPS Pressure switch number: Don’t show me this message again.
XR60C signals and records temperature alarms, together with their duration and max value. In case of applications in industrial environments, the use of mains filters our mod. Enter text from picture: After defrost, there is a timed fan delay allowing for drip time, set by means of the “Fnd”.
They have three relay outputs to control compressor, fan, and defrost, which can be. In case of failure or faulty operation send the instrument back to the distributor or to “Dixell.